HACH SC100 MANUAL PDF
View and Download Hach Sc E instrument manual online. Analysis System. Sc E Measuring Instruments pdf manual download. Hach Sc Controller User Manual – Download as PDF File .pdf), Text File .txt) or read online. Hach Sc Controller User Manual . Hach Company, – .. 37 W with 25 W sensor/network card load 24 VDCpowered sc model: 24 VDC.
|Published (Last):||17 October 2007|
|PDF File Size:||17.15 Mb|
|ePub File Size:||2.58 Mb|
|Price:||Free* [*Free Regsitration Required]|
Hex nut, M6 4Cat. Flat washer 4Cat.
The passcode is factory set to sc the five digits must be followed by a space to remove the trailing asterisk. Wire gauge less than 18 AWG is not recommended.
Pay attention to all danger and caution statements. Close and secure the cover. Wet Verification System Startup 5. Pipe vertical or horizontal 5.
Index Index Numerics 90 Degree Detector Page of 72 Go. Verify Modbus master Parity setting is set to None. The time is available only in hour military format. Warranty, liability and complaints 7. Tell us about it. The display will flash on startup, when a sensor error has occurred, when the hold outputs function has been activated, and hahc a sensor is being calibrated.
Section 5 System Startup 5. Section 1 Specifications Specifications are subject to change without notice. Hex nut 4Cat.
Pass the cable through conduit and a conduit hub or cs100 strain relief fitting and an available access hole in the controller enclosure. Don’t show me this message again. See Replacement Parts and Accessories on page 55 for ordering information.
Failure to do so could result in serious injury to the operator or damage to the equipment. The controller contains three relays designed for use with high voltage Danger: This is the default setting. Mounting bracket, controller, Cat.
Hach-Lange SC : Owner manual
The non-sensor end is connected to the controller as described in section 3. Page 21 Installation 3. Page 43 Section 6 Replacement Parts 6. Page 26 Installation 3. Comments to this Manuals Your Name.
Troubleshooting Troubleshooting Table 11 presents sensor warnings displayed in maunal Event Log, possible causes, and corrective actions. Got it, continue to print.
Hach-Lange SC 100 User Manual
Appendix A Modbus Protocol A. Page 58 Index All configuration settings and calibrations can be changed. The turbidimeter sensor must be mounted within six feet of the controller unless an extension cable is used. Page 33 Operation 4. Figure 22 Front of the Controller 1 sc 5 2 6 3 7 4 1.
Supplied maunal hardware is shown in Figure 6, Figure 7, and Figure 8. Restore sample flow to the instrument.
Page 50 Modbus Protocol Table 9: Failure to do so could result in serious injury to the operator or damage to the equipment.